Your shopping cart is empty!
0 reviews / Write a review
Tags: PH Detector, PH electrode, PH Tester, PH Composite, PH Electrode Sensor, PH Detection
Description: Fundamentally, a pH meter consists of a voltmeter attached to a pH-responsive electrode and a reference (unvarying) electrode. The pH-responsive electrode is usually glass, and the reference is usually mercury–mercurous chloride (calomel) electrode, although a silver-silver chloride electrode is sometimes used.pH probes measure pH by estimating the voltage or potential difference of the solution in which it is dipped. By measuring potential difference, hydrogen ion concentration can be calculated using the Nernst equation which gives the relationship between Hydrogen ion concentration and Voltage or Potential.This pH electrode has a single cylinder that allows direct connection to the input terminal of a pH meter, controller, or any pH device which has a BNC terminal. The pH electrode probe is accurate and reliable that can give almost instantaneous readings, helping you to check the quality of water momentarily.
Features:Model: E-201-C-9PH Range: 0 ~ 14pHTemperature Range: 0°C ~ 60°CZero Points: 7 ± 1 pHInput Impedance: ≤250MΩRepeatability: ≤0.017pHPort: BNC
Instruction Guides:1. When the electrode is going to use for the first time, the electrode bulb should be immersed in a 3.3N potassium chloride solution for two hours to activate.2. It should be washed in distilled water and blots by filter paper to prevent impurities. The electrode bulb and the liquid junction should be immersed in the test solution at the same time.3. No need to add a reference solution for pH composite electrode4. The electrode should avoid long-term immersion in distilled water or protein or acid chloride solution and prevent contact with silicone grease.
Package Include:1 x E-201 PH Composite Electrode
Your B2B Supplying Partner in Mainland China
Using Vast Majority of International Couriers
Always Pay with Trust by PayPal Express Checkout